CORAIL® Total Hip System
CORAIL® Total Hip System
With more than 2 million stems1 provided for patients worldwide and thirty years of clinical heritage, the CORAIL® Total Hip System now has the most extensive experience with a hydroxyapatite (HA) coated stem. Combining basic design features including shape, surface finish and extensive hydroxyapatite coating, with a simple compaction broach-only surgical technique, the CORAIL® Total Hip System has demonstrated reproducible results and long-term biomechanical joint restoration.
Features & Benefits
Collared Stem Option
Collared stems are available in both standard and coxa vara offset. The collar is designed to provide advantages in the early stability of the implant, allowing for earlier post-operative weight bearing, protection against subsidence, and a positive dispersion of the vertical forces via the collar into the medial calcar.2,3
The Stepped Geometry of the CORAIL Stem is designed to minimize shear forces and maximize compression loading in host cancellous bone.
Multiple Offset Options
The CORAIL Stem offers standard and high offset options which can enable femoral offset restoration and soft tissue tensioning. The CORAIL Stem also offers a coxa vara neck option which provides an additional option for varus neck angled patients.
The CORAIL Stem utilizes compaction broaching, aimed to preserve the blood supply to promote healing and growth of bone around the implant. 97% Implant Survivorship has been documented in 5.456 cases at 15 years.4
DePuy Synthes Hip Solutions NJR Results
CORAIL NJR Analysis of Collared and Collarless
CORAIL PINNACLE 3x Stronger Together
1. TSM Report, CORAIL WW implantations YTD, 1986 - 2017.
2. H. Strom, O. Nilsson, J. Milbrink, et al. The effect of early weight bearing on migration pattern of the uncemented CLS stem in total hip arthroplasty. J Arthroplasty, 22 (2007), p. 1122
3. Demey G, Fary C, Lustig S, Neyret P, Aït si Selmi T. Does a Collar Improve the Immediate Stability of Uncemented Femoral Hip Stems in Total Hip Arthroplasty? A Bilateral Comparative Cadaver Study. J Arthroplasty 26 (2011), No.8, p. 1549
4. Hallan G, Lie SA, Furnes O, Engesaeter LB, Vollset SE, Havelin L. Medium and long-term performance of 11,516 uncemented primary femoral stems from the Norwegian arthroplasty register. J. Bone Joint Surg. 2007; 89-B: 1574-1580
For product details such as indications, contraindications, warnings and precautions please consult the IFU.
For additional information, see also: www.corailpinnacle.net/corail
© Johnson & Johnson Medical Limited 2021. All Rights Reserved.
This site is published by the DePuy Synthes Companies, which are solely responsible for its content.
This site is for visitors from United Kingdom.
The third party trademarks used herein are trademarks of their respective owners.
Please refer always to the Instructions for Use / Package Insert that come with the device for the most current and complete instructions.